Segmented detectors in scanning transmission electron microscopy (STEM) have been rapidly progressed, and recently have reached to the pixel array detectors (PADs) enabling to obtain whole information in reciprocal space during probe scanning . One of the coauthor Ikuta also had developed a squarely arranged PAD for STEM, which consists of 8×8 detection portions . As an application using the PAD in STEM is phase retrieval such as the ptychographic microscopy . This iterative technique is capable to represent phase images, however, in which specimen thickness has to be enough thin to be as the weak phase objects. In order to overcome this restriction, we have proposed the novel STEM phase retrieval technique using annularly-shaped pixel array detector (A-PAD) combined with quasi-Bessel beam . In this paper, we demonstrate applicability of our method, called Phase Retrieval for Thick Specimens (PRETS), by using a SrTiO3  single crystal.
Figure 1 shows a schematic illustration of the proposed phase retrieval method in STEM. By using an annular aperture to form hollow-cone-shaped probe, its intensity distribution is elongated along the optical axis , as well as in Bessel beam , resulting in applicability to thick specimens. In this case, however, the depth of focus (DOF) is limited finitely due to the finite width of the annular slit. This means that such probe does not correspond to the Bessel beam providing infinite DOF but should be called as the quasi-Bessel beam. The annular aperture has the other effects to restrict the distribution of the electrons going forward to the detection plane, as shown in Fig. 1. This leads to that, for bright-field (BF) imaging, the pixelated detectors are permitted to be located just at the annular-shaped region where the direct beam illuminated. This is the reason why we have adopted the annularly arrayed pixel detectors. Each detectors in the A-PAD yields BF-STEM images containing different information of specimen. From these components, the phase information can be extracted and reconstructed by a dedicated Fourier filter having an acentric annular shape and a summation of the filtered components.
Figure 2 shows the developed A-PAD apparatus. As shown in Fig. 2(a), this system contains 31 detectors in which 24 channels (#0-23) are used for the phase retrieval and the others for normal BF imaging. Each channel consists of a bare optical fiber, one end of which is coated directly by fluorescent P-47 powders (Fig. 2(b)). The other ends of the fibers are connected to a multianode photomultiplier tube (PMT). These fibers transfer the photons converted from electrons on the detectors. Eventually, output signals from the PMT are fed into an image processing computer. The detector is mounted on the x-y position adjustable stage, as shown in Fig. 2(d), which was assembled in the STEM column.
Figure 3 shows phase maps of a SrTiO3  single crystal, for comparison in terms of thickness effect, obtained by means of the multi-slice simulations. The phase in the TEM corresponding to imaginary part of the exit wave is strongly affected by increment of specimen thickness, as shown in Fig. 3(b). In contrast, phase retrieved by the proposed method can represent directly atomic structures without the contrast reversal even at 15 nm in thickness. This clearly proves the effectiveness of our novel technique PRETS.
 H. Yang, et al., Ultramicrosc. 151 (2015) 232
 M. Taya, et al., Rev. Sci. Inst. 78 (2007) 083705
 M. J. Jumphry, et al., Nat. Comm., 3 (2012) 730
 T. Kawasaki, et al., Ultramicrosc. 110 (2010) 1332
 V. Grillo, et al., Phys. Rev. X, 4 (2014) 011013
To cite this abstract:Tadahiro Kawasaki, Takafumi Ishida, Tetsuji Kodama, Takayoshi Tanji, Takashi Ikuta; STEM phase retrieval method for thick specimen by using quasi-Bessel beam. The 16th European Microscopy Congress, Lyon, France. https://emc-proceedings.com/abstract/stem-phase-retrieval-method-for-thick-specimen-by-using-quasi-bessel-beam/. Accessed: April 21, 2019
EMC Abstracts - https://emc-proceedings.com/abstract/stem-phase-retrieval-method-for-thick-specimen-by-using-quasi-bessel-beam/